[Date Prev][Date Next][Thread Prev][Thread Next][Date Index][Thread Index]
Re: Tektronix 535 scope for coiling use
- To: tesla@xxxxxxxxxx
- Subject: Re: Tektronix 535 scope for coiling use
- From: "Tesla list" <tesla@xxxxxxxxxx>
- Date: Fri, 08 Apr 2005 21:13:03 -0600
- Delivered-to: email@example.com
- Delivered-to: firstname.lastname@example.org
- In-reply-to: <email@example.com>
- Old-return-path: <firstname.lastname@example.org>
- References: <email@example.com>
- Resent-date: Fri, 8 Apr 2005 21:15:10 -0600 (MDT)
- Resent-from: tesla@xxxxxxxxxx
- Resent-message-id: <RRcjhD.A.fN.Lh0VCB@poodle>
- Resent-sender: tesla-request@xxxxxxxxxx
Original poster: Terry Fritz <teslalist@xxxxxxxxxxxxxxxxxxxxxxx>
Those old Tek scope are pretty tough!! Be sure to work the switches a ton
of times since they are self cleaning types. Simple parts are easy to get
and the parts or well documented... But it just usually takes a little
TLC. to bring them back to normal.
At 08:42 PM 4/8/2005, you wrote:
I picked up that old scope I asked the list about, when at his place
before leaving, it had a good trace on both channels (it is a dual trace
scope) when I got home, it did not work, I traced the problem to a bad
Pot, which I may have located a replacement in iowa...
I needed a scope, but I am starting to think I need to get a scope to use
to help fix this one next time it breaks, and I should get a "parts scope"
or two of this model (535A), so I will have parts.
So if anyone has some old tek scope parts, or a dead 535A that they do not
want anymore, let me know what you would want for them...